- 2.0mp 1600x1200 Kodak KAI-2020 sensor
- USB 2.0 High-Speed, 16-bit output
- Dual read rates of 800kHz and 8MHz
- Electronic shutter speeds to 100 microseconds
- Air cooling to 45C below ambient
- Available 5 or 8-position color filter wheel (CFW)
- Available Integrated Guider Port (IGP)
The QSI 620 model camera employs a 2 megapixel Kodak interline transfer CCD image sensor with microlens technology. The high quantum efficiency, wide dynamic range and low noise performance make the QSI 620 ideally suited to a broad range of demanding applications.
600 Series cameras offer dual read rates optimized for a wide range of medical, scientific and astronomical applications. The High Quality mode, which is about 50% faster than the 500 Series, is designed for applications where achieving the highest possible Signal to Noise Ratio is paramount. High Speed mode of approximately 8 million pixels per second supports applications requiring a higher frame rate. This allows the new 8.3mp 683 to download a full-frame bin 1x1 16-bit images in slightly more than one second. The QSI 620 and 640 with interline transfer sensors, allow ROI reads at multiple frames per second.
QSI 600 Series cameras have a new cooling subsystem with a custom 2-stage TEC that yields regulated cooling of greater than 45C below the ambient temperature. This allows 600 Series cameras to achieve a regulated temperature of -20C even in typical office/lab environments or in embedded systems. Astronomers also benefit, especially those in the hot southwest and humid southeast where night time temperatures may stay above 80F.
QSI 600 Series cameras have an available 5 or 8-position Color Filter Wheel. Either filter wheel offers automated filter selection through a wide variety of camera control applications and QSI's Windows COM and Linux API's.
QSI cooled CCD cameras are designed to produce high quality images with extremely wide dynamic range, excellent linearity and exceptionally low noise.
The QSI 600 Series builds on the foundation of the popular QSI 500 Series. Dual read rates of up to 8 MHz with high-speed USB 2.0 and full 16-bit output allow 600 Series cameras to produce high quality images with high frame rates, extremely wide dynamic range, excellent linearity and exceptionally low noise. Cooling on the 600 Series is achieved with a custom 2-stage TEC supporting regulated cooling to greater than 45C, or greater than 50C with the option Liquid Heat Exchanger.
The QSI 600 Series has seven different models employing a comprehensive range of monochrome and single-shot color scientific grade CCDs up to 8.3mp. A variety of options and accessories are available to meet your specific medical, astronomical or industrial imaging objectives.
The superb imaging performance of 600 Series cameras is wrapped in an attractive, compact design with outstanding power efficiency and investment protecting upgradeability. The refined engineering and impressive fit, finish and attention to detail will surpass your highest expectations.
The Benefits of Scientific Imaging Performance
QSI's scientific-grade CCD cameras deliver image quality equal to or better than cameras costing thousands more. The lower noise and other benefits of a QSI camera allow you to capture better data in less time. What does scientific imaging performance really mean? There are four key traits that distinguish a scientific camera:
- High sensitivity
- Lower Noise
- Linear response to light
- Precisely regulated cooling
How sensitive a CCD it to light is expressed as its Quantum Efficiency (QE). Higher QE means you can collect more photons in any given amount of time giving you better looking images with a higher Signal to Noise Ratio (SNR). High sensitivity is almost purely a function of the CCD. Just about any camera using the same CCD will achieve a similar sensitivity to light.
However, even with the same CCD, the Signal to Noise Ratio between two different cameras can be dramatically different especially in the fainter, lower signal areas of an image. These lower signal areas, such as the fringes of nebulae or dust lanes in a galaxy, often comprise some of the most interesting parts of astronomical images. Maximizing the SNR in these dim areas is the key to revealing those otherwise hidden details.
Lower Noise = Higher Dynamic Range
Noise in a CCD camera is uncertainty in the signal you’re trying to measure. The lower the noise, the greater the dynamic range possible with any given CCD. There are several types of noise that contribute to the overall noise in a CCD image.
CCD images are exquisitely sensitive to any noise introduced by the camera’s electronics. Mitigating the electronic interference from the digital electronics and switching power supplies is one of the many challenges in designing and building a scientific grade CCD camera. Any noise introduced by the camera is relatively easy to see by analyzing bias frames. A Bias Frame is a zero-length (dark) exposure which captures the slightly varying “floor value” from pixel to pixel (the bias) plus any noise added during the process of reading the image from the CCD. By subtracting two bias frames, you can remove the underlying differences in the pixel bias leaving just the noise added to the image when it was read from the CCD. This subtraction of two bias frames is called a Read Noise Frame.
Complicating the story is that some types of noise are tougher to deal with than others. For instance, truly random noise that displays a Gaussian distribution can be reduced by combining images. The graph to the right shows the histogram of a bias frame from a QSI camera overlaid over histograms from two other cameras using the same Kodak CCD. The histograms have been normalized for easy visual comparison. The histogram from the QSI camera displays a nearly perfect Gaussian distribution. The familiarbell curve resembles a parabola when displayed with a log scale. Standard deviation (σ) is shown for each histogram.
The parabolic shape of the histogram is important but you can't infer too much about the magnitude of the noise from the width of a histogram without knowing more about the cameras. Differences in width can be caused by higher noise or possibly just differences in a camera's gain settings.
Averaging together multiple bias frames can tell you more about the distribution of pixel values and make it easier to visually identify any non-Gaussian noise in the images.
To the left is a histogram comparing the average of 10 bias frames from a QSI 520i to a camera from another manufacturer with the same sensor taken at the same -10C temperature. The graph has the same scale as the one above for the single bias frames.
Note that both histograms are now narrower than the histograms for the single frames. This is a result of the noise being reduced in the combined frames because the random (Gaussian) noise tends to cancel out getting you closer to the "true" value of the underlying data. Also note however that the red histogram trails off to the right no longer resembling a parabola like the blue histogram from the QSI camera. This non-Gaussian noise typically has components of both pattern noise and stray pixels that fall outside the target Gaussian distribution. Pattern noise in particular will not be reduced by combinging frames. Instead this type of noise combines just like signal reinforcing itself and reducing the signal to noise ratio in your final images.
Bias frame histograms, however, can’t tell you the full story. The width and shape of the histogram shows the magnitude and overall distribution of the noise, but it tells you little about the structure of that noise. To do that requires another tool, the Fourier Transform.
A Fourier Transform is a mathematical function that converts a signal (any signal) into a collection of sine waves which together exactly represent the original signal. For images, we’re interested in 2-dimensional Fourier Transforms which convert an image from the Spatial Domain, rows and columns in a digital picture, into an equivalent Frequency Domain image. A Fast Fourier Transform (FFT) is a special algorithm that runs much faster on computers when starting with square images where each side has a length equal to a power of 2 (e.g 256, 512, 1024).
To analyze the noise from a CCD camera, you create a Read Noise Frame as described above. By subtracting two bias frames, you end up with just the noise introduced by reading the image from the CCD. An FFT of that Read Noise Frame will reveal any pattern noise introduced by the camera.
An FFT of random noise displaying a Gaussian distribution will have a smooth dark background with a single bright pixel in the center. To the left is an FFT of a Read Noise Frame from a QSI 500 Series camera. The FFT of the QSI Read noise frame reveals no pattern noise and is, in fact, virtually indistinguishable from an FFT of mathematically generated Gaussian noise.
To the right is the FFT of a Read Noise Frame from a camera that introduces non-Gaussian, periodic noise into the image. Note the series of vertical lines running through the FFT and the three bright dots near the center. Those vertical lines in the FFT are the result of horizontal noise of various frequencies in the original image.
Periodic noise will not be reduced when combining multiple images; instead it is reinforced, just like signal. The net effect of such noise is to raise the noise floor, reducing the signal to noise ratio and hiding the faint details in your images. When many such images are combined, periodic noise may show up as visible patterns, such as faint horizontal or vertical lines, running through your images.
Roll your mouse over the image to the right to see an example from a camera that displays even stronger pattern noise.
The images below of a standard optical target demonstrate one of the benefits of low noise and a Gaussian distribution. The images were captured with a QSI 516 camera in a very dark room. A single image closely resembles a bias frame. The target is virtually indistinguishable from the background. Its Signal to Noise Ratio is nearly zero. With an average combine of 50 frames, light and dark areas are clearly visible. With 200 frames, the target pattern is easily identifiable. Note that even after combining 200 frames, the difference between the dark and bright pixels is only 0.45 ADUs on average.
Lower noise displaying a Gaussian distribution allows you to reveal
subtle details in your targets that would otherwise be lost
Average of 50 Images
Average of 200 Images
Accurately recording the relative amount of light between different objects requires a camera that delivers a linear response to varying levels of light. A CCD camera should record twice the photon flux from an object that is twice as bright. Cameras that use CCDs with anti-blooming protection will start to show a drop in linearity as the pixels begin to approach a high percentage of full well capacity, but should be nearly linear at lower signal levels.
The log scale graph above shows the response to light of a QSI 520i which uses an interline transfer CCD with anti-blooming protection. the log scale of the graph amplifies any discrepencies from linear response especially at the lowest signal levels. The data points are linear to less then 1% all the way down to the lowest signal level, well within the specification of the CCD. Lower noise is especially critical in maintaining linearity at these lower signal levels.
High linearity is obviously critical for scientific applications such as accurate photometry, but it's just as important for producing beautiful pictures of deep sky objects. A linear response to light and lower noise allows you to accurately capture the subtle details of your targets in less time. To achieve the same signal to noise ratio in the faintest areas of your images with a higher noise camera can require hours more exposure time. Turning that around, a lower noise camera will yield great looking images with a higher signal to noise ratio in much less time.
Precisely Regulated Cooling
Every image taken with a CCD camera has two distinct types of “signal”. A CCD records light striking it from the objects in your field of view, and it records heat from the sensor itself. Cooling the CCD reduces the contribution of this “thermal” current. With most Kodak CCDs, the amount of thermal current is cut in half for every roughly 6°C drop in the temperature of the CCD. So, dropping the temperature of the CCD by 36°C reduces the rate that thermal current builds by over 98%.
The rate that thermal current builds at any given temperature is very predictable and repeatable. That’s why dark frame subtraction is so effective. Subtracting an average or median combine of multiple dark frames taken at the same temperature as your light frames will remove the thermal current component (and bias) from your light frames. However, if the temperature varies either during the capturing of your light frames or dark frames by even a few tenths of a degree, dark frame subtraction becomes much less effective.
The graph above plots the temperature of the CCD in a QSI 500 Series camera cooling from an ambient temperature of 13.5°C (56°F) down to -20°C. The drop of a little over 33°C below the ambient temperature took just over six minutes. QSI 500 Series cameras will generally cool the CCD to approximately 38°C below the ambient temperature and then regulate the temperature to within 0.1°C.This fine control of the temperature allows for very precise and very repeatable dark frame subtraction. 600 Series cameras have increased cooling capacity to approximately 45°C below the ambient temperature.
High sensitivity, low noise, a linear response to light and precisely regulated cooling are four key traits of a scientific camera. All are required in order to maximize the dynamic range of your images. Given a constant signal, lower noise increases the Signal to Noise Ratio and dynamic range, allowing you to produce better images in less time. Capturing good looking images of bright targets is relatively easy, but if you want great results in the dimmer areas of your targets, achieving the same signal to noise ratio with a higher noise camera can require hours of additional dark time.
All significant performance characteristics of QSI cameras, includingLinearity, Read Noise, and Photon Transfer (Gain), are tested and documented as the final manufacturing step. Each camera's timing and voltages are carefully set to ensure maximum Charge Transfer Efficiency, and to minimize charge injection and other secondary noise sources.
QSI cameras have many other benefits including small size, minimal backfocus, available internal color filter wheel, low power consumption and exceptional build quality.
C-Mount Guide Camera Adapter – works with SX Lodestar
T-Mount Guide Camera Adapter – works with Orion SSAG
1.25” Filter Guide Camera Adapter – works with SBIG ST-i
|Name||QSI - 620ws 2.0mp Cooled CCD Camera w/ Mechanical Shutter & Color Filter Wheel|
|Custom Stock Status||This item is special order. Delivery times will vary. Place your order today and your card will not be charged until the item ships.|
|CCD Sensor||Kodak - KAI-2020|